N/APosted on - 12/12/2011
I received an error after I tried to set up a thermocouple task on PXIe-4353.
This device uses a CJC channel on another mechanism and is connected to other devices as well.
If you have any idea how to deal with this issue, please help me out.
DAQ Assistant Error
Error -201426 occurred at DAQ Assistant
Possible reason (s):
One or more devices do not support multidevice tasks.
Help! DAQ Assistant Error 201426
The error message itself stated the reason why this error appears. “One or more devices do not support multidevice tasks. “ That means the device do not support multidevice tasks.
The error message also stated the devices that are affected by the error. For you to solve this problem, you need to replace the device that is listed to the error message with a device that supports multi device tasks, this are the product which will support multi device task.
1. S Series devices
2. X Series data acquisition modules
3. C Series modules
4. SC Express data acquisition modules